Volume 8 Supplement 1
Construction of web-based remote diagnosis system using virtual slide for routine pathology slides of the rural hospital in Japan
- Ichiro Mori†1Email author,
- Takashi Ozaki†2,
- Yasuteru Muragaki†2,
- Takatoshi Ibata†3,
- Hiroshi Ueda†3,
- Toshihito Shinagawa†4 and
- Yoshiyuki Osamura†1
© Mori et al; licensee BioMed Central Ltd. 2013
Published: 30 September 2013
In recent years, Japan has been short of pathologists. There are about 2,000 board-certified pathologists against 130 million people. Most pathologists are working in the urban area and only few are available in rural regions. Telepathology in Japan developed under these circumstances, mainly for intraoperative frozen section diagnosis where no pathologist is available. We have been performing frozen section telepathology for more than 10 years, and achieved fairly good results .
In this work, we constructed a web-based remote diagnosis system using virtual slide targeting routine pathology slides.
Material and methods
In some difficult cases, we discussed with pathologist who visits the hospital once a week, and compared the diagnosis between VS and microscope.
Results and discussion
We started this system in July 2011, and continue to diagnose about 80 cases each month resulting in more than 800 primary diagnosis. The VS images were clear enough and we could diagnose most of the routine HE slides with no incident. No image degradation happened due to data transfer. Because VS are stored in the server and connected to the web, we can manage our time and location for diagnosis easily. I have also been able to view VS images from Europe using the internet at my hotel.
As a setting of VS scanner, we first tried to use 20x objective lens. But in some delicate cases, the images were frustrating for us. Since we are unable to predict the delicate cases, we fixed objective lens to 40x.
To check the discrepancies between direct microscope and remote VS diagnosis, we looked at some difficult cases and discussed together. The degrees of coincidence were fairly good.
The biggest issue was the data transfer speed. The response of the VS differed each time, sometimes it was really slow whilst at other times it was very fast. When comparing VS observation using broadband of 50-60 Mbps to a mobile telephone network of 1-5 Mbps, the difference was small. We also changed the internet connection of the hospital’s VS server which resolved the issue dramatically. There should be bottleneck somewhere between the VS server and the web this time.
One big trouble was misunderstanding of the slide direction of VS. We put multiple small specimens on one glass slide, and numbered them from the label side. Because our VS viewer does not show thumbnail images with labels, a pathologist cannot find which side in #1 on viewer. Technicians usually set glass slides to the VS scanner with the slide label on the right hand side, which makes the right side specimen #1. One day, a glass slide was accidently placed onto the scanner the opposite way round. There were 4 colon specimens on the VS, and the 2nd specimen from the right showed cancer. The specimen #2 was taken from the ascending colon, but actually it was the specimen #3 taken from the descending colon. Luckily we noticed this before the operation.
According to the Japanese Medical Practitioners Act, primary pathology diagnosis is designated as a medical act, and medical act must take place in medical institution. Using this web-based system, we can make primary diagnosis from anywhere in the world. But in accordance to the law, we kept the report as draft when made outside of the medical institution and it was later finished at the hospital.
Because we cannot access the hospital’s pathology reporting system directly from Tokyo, we cannot write the diagnosis directly. At least, this could help us because the pathology technicians can find simple mistakes in the report when they copy it.
There are many ideas reported for the usage of VS and Telepathology [2–5]. This time, we focused on the web-based primary diagnosis. We constructed this system as a kind of emergency response for the absence of a pathologist, and it proved to be useful. There still many issues that remain, like the way to write diagnosis report to the system, clarification of where responsibility lies, and how to handle the large operation specimens or cytology slides, etc. In Japan, we have a guideline for intraoperative frozen section telepathology diagnosis written by the Japanese Research Society of Telepathology and Pathology Informatics . We may require a guideline for web-based remote primary diagnosis using VS.
- Mori I, Wang X, Suzuma T, Yoshimura G, Sakurai T, Kakudo K: Assessment of telepathological frozen section diagnosis of sentinel lymph node of breast cancer. Virchows Archiv. 2003, 443 (3): 302-
- Wienert S, Beil M, Saeger K, Hufnagl P, Schrader T: Integration and acceleration of virtual microscopy as the key to successful implementation into the routine diagnostic process. Diagn Pathol. 2009, 4 (3): 1-8.
- Pantanowitz L, Valenstein PN, Evans AJ, Kaplan KJ, Pfeifer JD, Wilbur DC, Collins LC, Colgan TJ: Review of the current state of whole slide imaging in pathology. J Pathol Inform. 2011, 2: 36-10.4103/2153-3539.83746.PubMed CentralView ArticlePubMed
- Brachtel E, Yagi Y: Digital imaging in pathology - current applications and challenges. J Biophotonics. 2012, 5 (4): 327-35. 10.1002/jbio.201100103.View ArticlePubMed
- Isaacs M, Lennerz JK, Yates S, Clermont W, Rossi J, Pfeifer JD: Implementation of whole slide imaging in surgical pathology: A value added approach. J Pathol Inform. 2011, 2: 39.
- Tsuchihashi Y, Sawai T: Establishing Guidelines for Practical Telepathology in Japan. Gan No Rinsho. 2009, 51 (9): 721-725.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.